Radiation detected: Gamma Radiation. Range: 0- 20000 CPS.
Range and Unit are configurable.
Detector Probe: Smart Gamma Scintillation Probe. For detailed specifications of the above smart probes, refer to the following pages attached.
NORMAL & ACTIVE WINDOWS : The NORMAL WINDOWS (GREEN) glows during acquisition mode until the dose rate exceeds alarm, preset value. Once dose rate exceeds alarm preset value, then the NORMAL WINDOW turns OFF and the ACTIVE WINDOW (RED) blinks. Once the dose rate falls below the preset level, then
ETHERNET PORT : User can inter face with PC by using RADGRID software through Ethernet (RJ 45 CONNECTOR ).
POWER CONNECTOR : This is used to connect the supply to monitor through 5v adapter.
(i) if Reset mode is AUTO then it comes back to Normal mode.
(ii) if RESET mode is MANUAL, then it comes back to NORMAL mode only after RESET key is pressed on Touch Display.
DOSERATE WINDOW : This gives the visual display of the current dose rate in acquisition mode in the ‘unit’ selected. By default it shows in µR/hr unit.
5" LCD TOUCH DISPLAY :
RESET key: Reset key is used to reset the alarm, condition. In auto reset once dose rate falls below the preset alarm level, audio & visual alarms reset, automatically.
‘manual’ reset mode alarm active condition is
reset only when the user presses ‘Reset’ button.
ACK key: This key is used to ‘ACK’ the alarm, by the user (this function is explained, in detail, under operating instructions).
Electronic Unit :
Dimensions : 270L x 185H x 90D.
Weight : 4Kg approx.